2mm High Accuracy Hi-Target Total Station with 6000m Measuring Range
***Brief Description of Important Operations***
Distance Measurement: Electronic distance measuring (EDM) instrument is a major part of total station. Its range varies from 2.8 km to 4.2 km. The accuracy of measurement varies from 5 mm to 10 mm per km measurement. They are used with automatic target recognisers. The distance measured is always sloping distance from instrument to the object. Angle Measurements: The electronic theodolite part of total station is used for measuring vertical and horizontal angle. For measurement of horizontal angles any convenient direction may be taken as reference direction. For vertical angle measurement vertical upward (zenith) direction is taken as reference direction. The accuracy of angle measurement varies from 2 to 6 seconds.
This instrument is provided with an inbuilt microprocessor. The microprocessor averages multiple observations. With the help of slope distance and vertical and horizontal angles measured, when height of axis of instrument and targets are supplied, the microprocessor computes the horizontal distance and X, Y, Z coordinates. The processor is capable of applying temperature and pressure corrections to the measurements, if atmospheric temperature and pressures are supplied.
Display: Electronic display unit is capable of displaying various values when respective keys are pressed. The system is capable of displaying horizontal distance, vertical distance, horizontal and vertical angles, difference in elevations of two observed points and all the three coordinates of the observed points.
Electronic Book: Each point data can be stored in an electronic note book (like compact disc). The capacity of electronic note book varies from 2000 points to 4000 points data. Surveyor can unload the data stored in note book to computer and reuse the note book.
***Use of Total Station***
The instrument is mounted on a tripod and is levelled by operating levelling screws. Within a small range instrument is capable of adjusting itself to the level position. Then vertical and horizontal reference directions are indexed using onboard keys. It is possible to set required units for distance, temperature and pressure (FPS or SI). Surveyor can select measurement mode like fine, coarse, single or repeated.
When target is sighted, horizontal and vertical angles as well as sloping distances are measured and by pressing appropriate keys they are recorded along with point number. Heights of instrument and targets
can be keyed in after measuring them with tapes. Then processor computes various information about the point and displays on screen. This information is also stored in the electronic note book. At the end
of the day or whenever electronic note book is full, the information stored is downloaded to computers.
The point data downloaded to the computer can be used for further processing. There are software like auto civil and auto plotter clubbed with AutoCad which can be used for plotting contours at any specified interval and for plotting cross-section along any specified line.
***Advantages of Using Total Station***
The following are some of the major advantages of using total station over the conventional surveying instruments:
1. Field work is carried out very fast.
2. Accuracy of measurement is high.
3. Manual errors involved in reading and recording are eliminated.
4. Calculation of coordinates is very fast and accurate. Even corrections for temperature and pressure are automatically made.
5. Computers can be employed for map making and plotting contour and cross-sections. Contour intervals and scales can be changed in no time.
However, surveyor should check the working condition of the instruments before using. For this standard points may be located near survey office and before taking out instrument for field work, its working is checked by observing those standard points from the specified instrument station.
1.Angle measurement (Hz,V)
|Detecting system||Absolute encoder|
|Field of view||1°30′|
|Minimum focus distance||1.5m|
4.Distance measurement (IR)
|Single prism (good condition)||3000m|
|Measuring time (Repete/Tracking)||1s/0.5s|
|Range (white board reflectivity 90%)||600m|
|Internal memory||20000 points (SD card supported)|
|Interface||RS232, Mini-USB, USB|
|Operation system||Real-time operation system|
|Display||192×96 high-light LED screen, brightness adjustable in 3 level|
|Keyboard||2 sides alphanumeric backlit crystal keyboard|
|Type||Laser point, 4 brightness levels adjustable (Optical plummet is optional)|
|Accuracy||1mm at 1.5m instrument height|
|Meteorological parameter modification||Manually(T-P senser is optional)|
|Dust & Water proof||IP66|
|Battery type||BT30 Li-ion battery|
|Operating time||18 hours(continuous angle measurement every 30 seconds)|
|Measuring times||Approx. 12000 times|
|Weight||3.8kg with battery|
***Characters of HI-TARGET total station***
*Zenith point measurement without removing the handle
*Shortcut measurement key on the side, convenient for surveying
*Small and light